An artificial sphincter consists of three parts:
To treat urinary incontinence, the cuff is placed around the bladder neck so that when it is inflated, the urethra will close tightly. The pressure regulating balloon will be placed under the tissues of the lower abdomen. The balloon is filled with a liquid (occasionally, an iodine-based solution will be used for better visualization when x-ray procedures are performed). The control pump mechanism is placed in the labia for women, and in the scrotum for men. To use the sphincter, the person will compress (squeeze) the pump to divert fluid from the urethral cuff to the balloon. This action will allow the sphincter to relax so that the person can urinate. The cuff will then re-inflate on its own in 3 - 5 minutes .
© 2011 University of Maryland Medical Center (UMMC). All rights reserved.
UMMC is a member of the University of Maryland Medical System,
22 S. Greene Street, Baltimore, MD 21201. TDD: 1-800-735-2258 or 1.866.408.6885